Richter Cine Collimators and Test Projectors
©2016-2018 Jorge Diaz-Amador, all rights reserved
Updated: 9 November 2018
Above: Richter Cine R2 Autocollimator setup with 254mm Objective, LED Mini-Maglite, M2 Focal Plane Micrometer, ARRI PL Lens Standard on Model B Optical Bench.
Foreground items: Fiber Optic Light Guide Adapter, 60mm Collimator Objective, ARRI Standard and ARRI Bayonet Lens Standards.
This was the original CinemaTechnic collimator which we used for all our lens and camera service from 2000 through 2006. The unit pictured above was sold in 2009.
Richter Cine Equipment
Richter Cine was an American company, based in upstate New York, that manufactured autocollimators and lens test projectors specifically designed for use in testing photographic and particularly cinematographic optics.
The company was founded by Kenneth Richter, who was a cinematographer and a filmmaker specializing in travelogue films. The Richter Autocollimator system is Richter Cine’s best known product.
By the 1980’s the Richter Autocollimator became very widely used in the motion picture camera and lens industry. In 1985 Kenneth Richter was awarded an AMPAS Technical Award (Technical Oscar) for his Autocollimator.
Production of the autocollimators and the Lens Test Projector were wound down by the late 1990’s. Richter Aero Equipment ended operations by October 1998. After Kenneth Richter’s passing two years later, the last remaining old stock was sold off.
Today used Richter Cine Autocollimators are highly sought after in the second hand market. The represent by far the best value in lens testing equipment.
How Autocollimators Work
For photographic and cinematographic lens testing, autocollimators are primarily used to check focus. The autocollimator projects the image of a test reticle through the lens being tested and onto a simulated or real image plane. That image is reflected back through the test lens and the autocollimator and diverted to the eyepiece where it can be observed by the operator. With experience, the image quality of the lens being tested can also be judged.
Reconditioning and upgrades for Richter Cine Autocollimators
- CinemaTechnic have developed a series of services and upgrades to the vintage Richter Cine autocollimators. This service brings the vintage Richter units up to today’s standards of accuracy.
- Disassembly and cleaning of all components. Precision cleaning of optical components
- Worn parts are reconditioned or replaced with newly fabricated parts
- Flatness of objective seating surface and condition of internal thread on autocollimator block checked and if necessary re-machined
- Collimator block internal components precision calibrated for for exact and equal focus on focus reticle and grid reticle assuring accuracy of the instrument for focus calibration of lenses. Flange focal distance of Grid Reticle and Focus Reticle are each individually adjusted and checked using a larger autocollimator as reference. This ensures the accuracy of the R2 Collimator Block
- Squareness and alignment of components of collimator objective tubes checked and re-machined as needed. Loose fitting components are bonded in place with industrial adhesives while held in alignment fixtures.
- Worn threaded surfaces coated with Emralon to restore proper fit, or in cases of severe wear replaced with newly fabricated parts.
- Worn exterior surfaces stripped and re-anodized or coated with anticorrosion coatings where needed.
- Collimator block internal components precision calibrated for for exact and equal focus on focus reticle and grid reticle assuring accuracy of the instrument for focus calibration of lenses
- Objectives precision calibrated for matching focus ensuring interchangeability without re-calibration
- Focal Plane Micrometer (FPM) reconditioned and tested for accuracy. Option to replace the analog micrometer with a digital unit.
- CinemaTechnic upgrade for Richter FPM to allow quick and easy zeroing of micrometer thimble
- Complete system is tested for accuracy against our Moeller-Wedel autocollimator and with test lenses calibrated on the Zeiss K9 MTF Tester.
Pricing for the above service is by estimate. Without examining your Richter system we cannot quote a price on the upgrade / reconditioning service. As a rough guide, our basic pricing is below.
Basic pricing for Richter Cine calibration:
- Focus calibration of collimator objectives – $63-$75 each (includes element cleaning and dust blow-out)
- Focus calibration and alignment of collimator block – $225-$250
- Calibration of Richter M2 Focal Plane Micrometer – apx. $400 depending on condition. Calibration and Reconditioning of Richter M2 with new precision front surface mirror, mirror support and mirror bezel, and coating of mirror cage anti-reflex black (see photo below) Discontinued (due to excessive labor time) – we can supply parts (mirror, o-ring, etc) if you wish to do this upgrade yourself We recommend digital conversion instead.
- Conversion of Richter M2 to Digital Micrometer – details and pricing TBA
Looking for a good used Richter Autocollimator system?
CinemaTechnic has one rebuilt and calibrated Richter Collimator system available for sale (see photos below). The system has been reconditioned and upgraded as described in the preceding section. It is priced at a fraction of what new autocollimator systems (Chrosziel, Gecko-Cam, Duclos) are being sold for today. The package consists of:
Complete Autocollimator System (see photo above)
- Richter Cine R2 Autocollimator Block, rebuilt with CinemaTechnic upgrades
- Richter Cine 60mm Collimator Objective
- Richter Cine 130mm Collimator Objective
- Richter Cine 180mm Collimator Objective
- Richter Cine 254mm or 275mm Collimator Objective
- CinemaTechnic Fiber Optic Light Guide Adapter for R2 Autocollimator
- Richter ARRI PL mount Lens Mount
- Custom Canon EF mount Lens Mount
- Richer M2M Focal Plane Micrometer with Starret 468M micrometer head, .002mm accuracy, new reflector with CinemaTechnic upgrades.
- Richter Model B Optical Lens Test Bench with 36″ rods
- Richter Cine Lens Support for OTB
- Halogen Fiber-Optic Illuminator with adjustable intensity, fiber-optic light guide.
- Set of instructions
- Calibration warranted for 1 year
- Phone/email tech support for 30 days included, one-on-one training available at additional cost
Compare to Duclos Lenses collimator system at $19,695.- or Gecko-Cam’s collimator system at € 31,700.- (currently $41,158.- plus shipping from Germany).
Complete system as above: SOLD
The complete system is well suited for serious users like camera/lens rental houses, production companies with their own equipment and in-house technical, and independent camera/lens technicians.
The complete system is also the best choice for camera owners needing to calibrate FFD on cameras such as ARRI Alexa / Amira that use shims to calibrate FFD. A variable focus test lens is required for camera testing. We can provide advice on choosing a test lens and can provide modification and calibration of test lenses.
Richter Cine Individual Components
We have some Richter Cine and Moeller-Wedel components available at the moment but do not have a complete collimator system available.
Lens Mounts (Standards)
ARRI PL Mount with Stainless Steel flange and ring for Richter Cine (CinemaTechnic custom) $875.-
List Price $638.-
Pre-order Price $499.- with 20% deposit
If you need a still camera mount other than Canon EF please contact us via email. We may be able to accommodate your request for the pre-order price.
NOTE: You will need a Flange Focal Depth gage set to calibrate the Focal Plane Micrometer and an Optical Flat to check calibration on the collimator optics. These items will be at additional cost and quoted based on your requirements (number of lens mounts you want to be able to calibrate, etc.)
Richter Cine compared to more costly modern systems
- A method to precisely calibrate the reflex blocks (0.015mm accuracy)
- A method to precisely calibrate focus on the collimator objectives (1/2000th focal length accuracy)
- A procedure to refurbish the M2 Focal Plane Micrometers including new reflectors
- Improvements in light source
- New mounts that were never available from Richter (Canon EF, Sony E)
- An upgrade to the IMS mounts system that can be easily removed for compatibility with Sony E mount and C-mount.
IMS Mount Upgrade
The IMS (Interchangeable Mount System) was developed by P+S Technik of Munich, Germany to be an open standard sub-mount system for both test equipment and interchangeable mount cameras. The IMS system is used on the P+S Technik Lens Checker (lens test projector) and the Gecko-Cam Autocollimator and Lens Test Projector
Available lens mounts in the IMS system: ARRI PL, B4, BNCR, C-mount, Canon EF, Canon FD, Nikon F, Leica M, Leica-R, Panavision PV-35.
CinemaTechnic can also manufacture custom IMS mounts We have built ARRI PL, BNCR, Canon FD, Leica-R and IMAX IMS mounts with stainless steel rear flanges.
CinemaTechnic Digital Focal Plane Micrometer
- Super precision digital Focal Plane Micrometer with interchangeable mount plate.
- 0.001 mm / 0.00005 in accuracy (one-thousandth millimeter, fifty millionths inch)
- Excellent image quality on reflex autocollimator
- Quickly set zero or pre-set to a distance (ex. 52.000 mm)
- Shown in photo with Richter standard mount plate
- Precision no-stress, no-damage clamping system for micrometer stem
- Compatible with ANY lens mount regardless of Flange Focal Distance
- Available for Richter Cine, Chrosziel, and other autocollimator systems
Options for Richter Cine Autocollimator (additional cost):
- CinemaTechnic IMS Mount Adapter for Richter Cine – compatible with M2 Focal Plane Micrometer and S2 Focal Plane Microscope. $775.-
- CinemaTechnic Focal Plane Micrometer (replaces M2) with digital micrometer (.001mm accuracy) and Richter flange, upgradable to IMS. Price by quotation only.
- CinemaTechnic Video Upgrade for Richter Cine Autocollimator – mounts a small c-mount camera (SD or HD) in place of the eyepiece. Greatly reduces uses fatigue. Price TBA.
- P+S Technik IMS mounts (Interchangeable Mount System). Available mounts: ARRI PL, B4, BNCR, Canon EF, Canon FD, Leica-M, Leica-R, Nikon F, Panavision PV-35. From €500.-
- CinemaTechnic Custom IMS mounts – most cine or still camera lens mounts with FFD over 25mm. From $345.-
- Additional NEW lens mounts with Richter Cine thread, custom made on request (C-Mount, Sony E, MFT, Leica-M, Leica-R, Canon FD, Nikon F, ARRI Standard, ARRI Bayonet, Aaton A50. From $545.-
- Original Richter Cine lens mounts (standards), good used condition. From $185.- (price depends on mount type)
- Additional Collimator Objectives (original Richter) 60mm, 130mm, 254mm. From $265.-
- Additional Collimator Objectives (built by CinemaTechnic, new production): 100mm, 400mm, 500mm. Price by quotation only.
- Back focus Master Lens (modified IBE Null) for checking autocollimator and testing/adjusting FFD on digital cameras
- Additional length rods for collimator bench
- FFD Calibration Kit for PL Mount
- FFD Calibration Kit for EF Mount
- Optical flat for autocollimator calibration check
- Digital video camera to replace autocollimator eyepiece (customer to provide monitor)
- Lens Support for collimator bench (for large heavy zoom lenses)
If you are interested in purchasing this system, or if want to purchase individual Richter Cine components, or want more information please contact us at:
The Richter Cine R2 Reflex Autocollimator
Richter Cine’s best known product is the R2 reflex autocollimator. The system was comprised of an autocollimator block (the reflex unit), a collimator objective, and light source. It was possible to see the image reflected by film in the aperture of the camera being tested when a lens and camera were tested together.
Richter Cine manufactured a series of interchangable collimator objectives for the R2, using a C mount thread for attachment to the collimator block.
Shorter focal length objectives (60, 130mm) were used to test shorter focal length lenses and longer focal length objectives (254 – 500mm) were used to test longer focal length lenses.
The following is a list of all Richter objectives known to be available. The most common are 60mm, 130mm and 254mm. Part numbers beyond 500mm are unknown and the 650mm and longer objectives were special-order items produced in very limited quantity.
Standard objectives up to 650mm are refractor objectives using only spherical lens elements. Objectives longer than 650mm are mirror lens objectives, using a combination of spherical mirrors (like an astronomical telescope) and spherical lens elements. The mirror lens design makes the physical length of those objectives much shorter than their focal length.
Richter Cine Collimator Objectives:
Focal Length | f# | Type | Test Lens Range | Richter Part #
- 60mm refractor objective 6 – 30mm #117
- 130mm f4.3 refractor objective 13 – 65mm #118 standard equipment for R2F
- 180mm f3.7 refractor objective 18- 90mm #119
- 254mm f4.9 refractor objective 25 – 127mm #120 standard equipment for R2A and R2B
- 275mm f8.3 refractor objective 27 – 138mm (part # not known, early model that was later discontinued)
- 500mm f9.6 refractor objective 50 – 250mm #121
Special order objectives:
- 650mm f10 refractor objective 65 – 325mm (very rare)
- 1000mm f11 Maksutov-Cassegrain catoptric objective, 200mm tube length 100 – 500mm
- 1250mm f10 Schmidt-Cassegrain catoptric objective, 275mm tube length 125 – 625mm
- 2000mm f10 Schmidt-Cassegrain catoptric objective, 425mm tube length 200 – 1000mm
NOTE: Catoptric objectives are “mirror lenses” like modern astronomical telescopes.
“The rule of thumb is, to obtain the optimum collimator test image, the range of any lens to be tested should have a focal length not less than 1/10th or not more than 1/2 the focal length of the objective.”
Source: Instructions for Use of the Richter Cine Equipment Reflex Autocollimator Model R-2 ©1961 Richter Cine Equipment
All text and photos ©2016-2018 Jorge Diaz-Amador. All text, images and other content on this site are original copyrighted works. CinemaTechnic and Jorge Diaz-Amador own exclusive rights to this content under the United States Copyright Act of 1976. Unauthorized use and/or duplication of this material without express and written permission from this website’s author and owner is strictly prohibited.
We are now actively monitoring for infringing websites that copy our content.
Last update: 11/08/2018